Subject: [IGSSTATION-3465]: WTZJ Radome LEIT installed together with Antenna LEIAR25
From: Rudi Stöger <[email-masked]>
To: [email-masked], Rudolf Stoeger <[email-masked]>
Date: Mon, 21 Dec 2009 11:03:33 +0100

******************************************************************************
IGS Station Mail          21 Dec 02:05:02 PST 2009      Message Number 3465
******************************************************************************

Author: Rudi Stoeger

Ladies and Gentlemen

My colleagues replaced the TRM29659.00 antenna
with a LEIAR25 antenna without a radome
in the WTZJ GPS+GLONASS Permanent Tracking
Station in the Geodetical Observatory
Wettzell, Germany, 2009-12-16T12:00Z.

On 2009-12-17T07:35Z my colleagues also installed
the radome LEIT.

With best regards
Rudi
     International GPS Service
     WTZJ Site Information Form
     See Instructions at:
       ftp://igscb.jpl.nasa.gov/pub/station/general/sitelog_instr.txt

0.   Form

     Prepared by (full name)  : Rudolf Stoeger
                              : Peter Franke
                              : Matthias Groeschel
     Date Prepared            : 2009-12-16
     Report Type              : UPDATE
     If Update:
      Previous Site Log       : wtzj_20080124.log
      Modified/Added Sections : 1, 4.3, 4.4, 5.6, 7.19,
                              : 7.20, 7.21, 7.22, 7.23,
                              : 7.24, 7.25, 7.26, 7.27,
                              : 7.28, 7.29, 7.30, 7.31,
                              : 7.32, 11, 13

1.   Site Identification of the GNSS Monument

     Site Name                : Wettzell / Germany
     Four Character ID        : WTZJ
     Monument Inscription     : Pillar 1203
     IERS DOMES Number        : 14201M012
     CDP Number               :
     Monument Description     : STEEL PLATE ON CONCRETE SURVEY TOWER
       Height of the Monument : 0.8
       Monument Foundation    : SCREW ANCHORED TO CONCRETE SURVEY TOWER
       Foundation Depth       :
     Marker Description       :
     Date Installed           : 2001-07-03T00:00Z
     Geologic Characteristic  : Bedrock
       Bedrock Type           : Cordirit-Silimanit-Gneiss
       Bedrock Condition      : Weathered
       Fracture Spacing       : 50-200 cm and over 200 cm
       Fault zones nearby     : NO
         Distance/activity    :
       Additional Information : The antenna is 7.5 m above
                              : the ground.
                              : On the same pillar 1203,
                              : DOMES No. 14201M012, also
                              : a 3SNAV R101 GPS+GLONASS Receiver,
                              : Four Character ID WTZG, was
                              : observing from 14-OCT-1998 until
                              : 04-SEP-1999.
                              : In addition and parallel a
                              : LEICA GRX1200GGPRO GPS+GLONASS Receiver,
                              : Four Character ID WTZR, IGS
                              : Reference Frame Station, on
                              : pillar 1202, Domes No. 14201M010,
                              : an ASHTECH Z-XII3T Metronome GPS
                              : Receiver, Four character ID WTZA,
                              : on pillar 1204, Domes No. 14201M013,
                              : a TPS E_GGD GPS+GLONASS Receiver,
                              : Four character ID WTZZ, on pillar
                              : 1205, Domes No. 14201M014, a 
                              : TPS NETG3 GPS+GLONASS Receiver,
                              : Four character ID WTZL, on
                              : pillar 1206, DOMES No. 14201M022
                              : and a SEPT POLARX2, Four Character
                              : ID WTZS, DOMES No. 14201M015
                              : are observing.
                              : Prior we observed here in the
                              : Fundamentalstation Wettzell with a
                              : TI-4100, a MiniMac 2816 AT, a
                              : Rogue SNR-800, our # 1200, Four Character
                              : ID WETT, Domes No. 14201M009,
                              : the 3SNAV R101 GPS+GLONASS
                              : Receiver, Four character ID WTZG,
                              : on pillar 1203, Domes No.
                              : 14201M012, and a TRIMBLE 4000 SSI,
                              : Four character ID WTZT.
                              : Further informations and effective
                              : dates see sections 5 and 7.

2.   Site Location Information

     City or Town             : Bad Koetzting
     State or Province        : Bavaria
     Country                  : Germany
     Tectonic Plate           : Eurasian
     Approximate Position
       X coordinate (m)       : 4075580.0194
       Y coordinate (m)       : 931855.2287
       Z coordinate (m)       : 4801568.1824
       Latitude (N is +)      : +490839.12
       Longitude (E is +)     : +0125244.15
       Elevation (m,ellips.)  : 00665.9
     Additional Information   :

3.   GNSS Receiver Information

3.1  Receiver Type            : JPS LEGACY
     Satellite System         : GPS+GLONASS
     Serial Number            : MT312211422
     Firmware Version         : EURO_3/2.1 JAN,03
     Elevation Cutoff Setting : 0
     Date Installed           : 2001-07-03T12:00Z
     Date Removed             : 2002-10-09T12:00Z
     Tempearture Stabiliz.    : 2.0
     Additional Information   :

3.2  Receiver Type            : JPS LEGACY
     Satellite System         : GPS+GLONASS
     Serial Number            : MT312211422
     Firmware Version         : EURO_3/2.2p2
     Elevation Cutoff Setting : 0
     Date Installed           : 2002-10-09T12:00Z
     Date Removed             : 2003-11-24T14:00Z
     Tempearture Stabiliz.    : 2.0
     Additional Information   :

3.3  Receiver Type            : JPS LEGACY
     Satellite System         : GPS+GLONASS
     Serial Number            : MT312211422
     Firmware Version         : EURO_3/2.3p3
     Elevation Cutoff Setting : 0
     Date Installed           : 2003-11-24T14:00Z
     Date Removed             : 2004-05-14T12:00Z
     Tempearture Stabiliz.    : 2.0
     Additional Information   :

3.4  Receiver Type            : JPS LEGACY
     Satellite System         : GPS+GLONASS
     Serial Number            : MT312211422
     Firmware Version         : Euro_3/2.3p4
     Elevation Cutoff Setting : 0
     Date Installed           : 2004-05-14T12:00Z
     Date Removed             : 2007-10-30T11:30Z
     Tempearture Stabiliz.    : 2.0
     Additional Information   : Additional Number
                              : LE 2808 on the
                              : type level

3.5  Receiver Type            : JPS LEGACY
     Satellite System         : GPS+GLONASS
     Serial Number            : MT312211422
     Firmware Version         : 2.6.0 OCT,24,2007 O
     Elevation Cutoff Setting : 0
     Date Installed           : 2007-10-30T11:30Z
     Date Removed             : 2008-01-24T08:44Z
     Tempearture Stabiliz.    : 2.0
     Additional Information   : Additional Number
                              : LE 2808 on the
                              : type level

3.6  Receiver Type            : JPS LEGACY
     Satellite System         : GPS+GLONASS
     Serial Number            : MT312211422
     Firmware Version         : 2.6.1 JAN,10,2008
     Elevation Cutoff Setting : 0
     Date Installed           : 2008-01-24T08:45Z
     Date Removed             : CCYY-MM-DDThh:mmZ
     Tempearture Stabiliz.    : 2.0
     Additional Information   : Additional Number
                              : LE 2808 on the
                              : type level

4.   GNSS Antenna Information

4.1  Antenna Type             : TRM29659.00     NONE
     Serial Number            : 0220180414
     Antenna Reference Point  : BPA
     Marker->ARP Up Ecc. (m)  : 000.0760
     Marker->ARP North Ecc(m) : 000.0000
     Marker->ARP East Ecc(m)  : 000.0000
     Alignment from True N    : 0
     Antenna Radome Type      : NONE
     Radome Serial Number     :
     Antenna Cable Type       : RG214
     Antenna Cable Length     : 30
     Date Installed           : 2001-07-03T12:00Z
     Date Removed             : 2002-04-11T09:30Z
     Additional Information   :

4.2  Antenna Type             : JPSREGANT_SD_E  NONE
     Serial Number            : RA0295
     Antenna Reference Point  : BPA
     Marker->ARP Up Ecc. (m)  : 000.0740
     Marker->ARP North Ecc(m) : 000.0000
     Marker->ARP East Ecc(m)  : 000.0000
     Alignment from True N    : 0
     Antenna Radome Type      : NONE
     Radome Serial Number     :
     Antenna Cable Type       : RG214
     Antenna Cable Length     : 30
     Date Installed           : 2002-04-11T09:30Z
     Date Removed             : 2005-08-02T09:00Z
     Additional Information   :

4.3  Antenna Type             : TRM29659.00     NONE
     Serial Number            : 0220175208
     Antenna Reference Point  : BPA
     Marker->ARP Up Ecc. (m)  : 000.0740
     Marker->ARP North Ecc(m) : 000.0000
     Marker->ARP East Ecc(m)  : 000.0000
     Alignment from True N    : 0
     Antenna Radome Type      : NONE
     Radome Serial Number     :
     Antenna Cable Type       : RG214
     Antenna Cable Length     : 30
     Date Installed           : 2005-08-02T09:00Z
     Date Removed             : 2009-12-16T11:59Z
     Additional Information   : Antenna calibrated absolute
                              : by GEO++ GmbH Garbsen,
                              : 2000-10-31.

4.4  Antenna Type             : LEIAR25         NONE
     Serial Number            : 08430008
     Antenna Reference Point  : BPA
     Marker->ARP Up Ecc. (m)  : 000.0740
     Marker->ARP North Ecc(m) : 000.0000
     Marker->ARP East Ecc(m)  : 000.0000
     Alignment from True N    : 0
     Antenna Radome Type      : NONE
     Radome Serial Number     :
     Antenna Cable Type       : RG214
     Antenna Cable Length     : 30
     Date Installed           : 2009-12-16T12:00Z
     Date Removed             : (CCYY-MM-DDThh:mmZ)
     Additional Information   : Antenna calibrated absolute
                              : by GEO++ GmbH Garbsen,
                              : 2008-11-20 together with
                              : radome LEIT.
                              : The radome LEIT is not installed.

4.x  Antenna Type             : (A20, from rcvr_ant.tab; see instructions)
     Serial Number            : (A*, but note the first A5 is used in SINEX)
     Antenna Reference Point  : (BPA/BCR/XXX from "antenna.gra"; see instr.)
     Marker->ARP Up Ecc. (m)  : (F8.4)
     Marker->ARP North Ecc(m) : (F8.4)
     Marker->ARP East Ecc(m)  : (F8.4)
     Alignment from True N    : (deg; + is clockwise/east)
     Antenna Radome Type      : (A4 from rcvr_ant.tab; see instructions)
     Radome Serial Number     :
     Antenna Cable Type       : (vendor & type number)
     Antenna Cable Length     : (m)
     Date Installed           : (CCYY-MM-DDThh:mmZ)
     Date Removed             : (CCYY-MM-DDThh:mmZ)
     Additional Information   : (multiple lines)

5.  Surveyed Local Ties

5.1  Tied Marker Name         : VLBI Ref Point
     Tied Marker Usage        : VLBI
     Tied Marker CDP Number   : 7224
     Tied Marker DOMES Number : 14201S004
     Differential Components from GNSS Marker to the tied monument (ITRS)
       dx (m)                 : -40.1314
       dy (m)                 : -119.9655
       dz (m)                 : +61.1925
     Accuracy (mm)            : 1.3
     Survey method            : TRIANGULATION
     Date Measured            : 2004-10-01T10:00Z
     Additional Information   :

5.2  Tied Marker Name         : SLR WLRS
     Tied Marker Usage        : SLR
     Tied Marker CDP Number   : 8834
     Tied Marker DOMES Number : 14201S018
     Differential Components from GNSS Marker to the tied monument (ITRS)
       dx (m)                 : -3.1580
       dy (m)                 : -69.7708
       dz (m)                 : +15.3895
     Accuracy (mm)            : 0.6
     Survey method            : TRIANGULATION
     Date Measured            : 2004-10-01T10:00Z
     Additional Information   :

5.3  Tied Marker Name         : TIGO VLBI
     Tied Marker Usage        : VLBI
     Tied Marker CDP Number   : 759324012
     Tied Marker DOMES Number : 14201S100
     Differential Components from GNSS Marker to the tied monument (ITRS)
       dx (m)                 : -8.2314
       dy (m)                 : -100.3115
       dz (m)                 : +15.1224
     Accuracy (mm)            : 0.9
     Survey method            : TRIANGULATION
     Date Measured            : 2004-10-01T10:00Z
     Additional Information   :

5.4  Tied Marker Name         : TIGO SLR
     Tied Marker Usage        : SLR
     Tied Marker CDP Number   : 75947901
     Tied Marker DOMES Number : 14201M200
     Differential Components from GNSS Marker to the tied monument (ITRS)
       dx (m)                 : +15.0227
       dy (m)                 : -122.2774
       dz (m)                 : -0.6106
     Accuracy (mm)            : 1.5
     Survey method            : TRIANGULATION
     Date Measured            : 2004-10-01T10:00Z
     Additional Information   :

5.5  Tied Marker Name         : 1200 ROGUE SNR-800 = WETT
     Tied Marker Usage        : GPS
     Tied Marker CDP Number   :
     Tied Marker DOMES Number : 14201M009
     Differential Components from GNSS Marker to the tied monument (ITRS)
       dx (m)                 : -1.4392
       dy (m)                 : -2.5501
       dz (m)                 : +1.8658
     Accuracy (mm)            : 1
     Survey method            : TRIANGULATION
     Date Measured            : 1995-02-13T15:00Z
     Additional Information   :

5.6  Tied Marker Name         : 1202 LEICA GRX1200GGPRO = WTZR
     Tied Marker Usage        : GPS+GLONASS
     Tied Marker CDP Number   :
     Tied Marker DOMES Number : 14201M010
     Differential Components from GNSS Marker to the tied monument (ITRS)
       dx (m)                 : +0.6658
       dy (m)                 : -1.5691
       dz (m)                 : -0.1282
     Accuracy (mm)            : 0.4
     Survey method            : TRIANGULATION
     Date Measured            : 2004-10-01T10:00Z
     Additional Information   : IGS Reference Frame Station

5.7  Tied Marker Name         : 1201 TRIMBLE 4000SSI = WTZT
     Tied Marker Usage        : GPS
     Tied Marker CDP Number   :
     Tied Marker DOMES Number : 14201M011
     Differential Components from GNSS Marker to the tied monument (ITRS)
       dx (m)                 : -2.4054
       dy (m)                 : +0.0527
       dz (m)                 : +2.0291
     Accuracy (mm)            : 0.4
     Survey method            : TRIANGULATION
     Date Measured            : 2004-10-01T10:00Z
     Additional Information   :

5.8  Tied Marker Name         : 1204 ASHTECH Z-XII3T = WTZA
     Tied Marker Usage        : GPS
     Tied Marker CDP Number   :
     Tied Marker DOMES Number : 14201M013
     Differential Components from GNSS Marker to the tied monument (ITRS)
       dx (m)                 : -1.5250
       dy (m)                 : -2.5706
       dz (m)                 : +1.7634
     Accuracy (mm)            : 0.4
     Survey method            : TRIANGULATION
     Date Measured            : 2004-10-01T10:00Z
     Additional Information   :

5.9  Tied Marker Name         : 1205 TPS E_GGD = WTZZ
     Tied Marker Usage        : GPS+GLONASS
     Tied Marker CDP Number   :
     Tied Marker DOMES Number : 14201M014
     Differential Components from GNSS Marker to the tied monument (ITRS)
       dx (m)                 : -0.4575
       dy (m)                 : -2.2556
       dz (m)                 : +0.7776
     Accuracy (mm)            : 0.4
     Survey method            : TRIANGULATION
     Date Measured            : 2004-10-01T10:00Z
     Additional Information   :

5.10 Tied Marker Name         : 1206 LEICA GRX1200PRO = WTZL
     Tied Marker Usage        : GPS
     Tied Marker CDP Number   :
     Tied Marker DOMES Number : 14201M022
     Differential Components from GNSS Marker to the tied monument (ITRS)
       dx (m)                 : -1.9769
       dy (m)                 : -1.2622
       dz (m)                 : +1.8975
     Accuracy (mm)            : 0.4
     Survey method            : TRIANGULATION
     Date Measured            : 2004-10-01T10:00Z
     Additional Information   :

5.11 Tied Marker Name         : SEPT POLARX2 = WTZS
     Tied Marker Usage        : GPS
     Tied Marker CDP Number   :
     Tied Marker DOMES Number : 14201M015
     Differential Components from GNSS Marker to the tied monument (ITRS)
       dx (m)                 :
       dy (m)                 :
       dz (m)                 :
     Accuracy (mm)            :
     Survey method            :
     Date Measured            :
     Additional Information   :

5.x  Tied Marker Name         :
     Tied Marker Usage        : (SLR/VLBI/LOCAL CONTROL/FOOTPRINT/etc)
     Tied Marker CDP Number   : (A4)
     Tied Marker DOMES Number : (A9)
     Differential Components from GNSS Marker to the tied monument (ITRS)
       dx (m)                 :
       dy (m)                 :
       dz (m)                 :
     Accuracy (mm)            : (mm)
     Survey method            : (GPS CAMPAIGN/TRILATERATION/TRIANGULATION/etc)
     Date Measured            : (CCYY-MM-DDThh:mmZ)
     Additional Information   : (multiple lines)

6.   Frequency Standard

6.1  Standard Type            : INTERNAL
       Input Frequency        :
       Effective Dates        : 2001-07-03/CCYY-MM-DD
       Notes                  :

6.x  Standard Type            : (INTERNAL or EXTERNAL H-MASER/CESIUM/etc)
       Input Frequency        : (if external)
       Effective Dates        : (CCYY-MM-DD/CCYY-MM-DD)
       Notes                  : (multiple lines)

7.   Collocation Information

7.1  Instrumentation Type     : SLR WLRS 8834
       Status                 : PERMANENT
       Effective Dates        : 1988-01-01/CCYY-MM-DD
       Notes                  :

7.2  Instrumentation Type     : SLR 7834
       Status                 : PERMANENT
       Effective Dates        : 1977-01-01/1991-12-31
       Notes                  :

7.3  Instrumentation Type     : SLR MTLRS 1
       Status                 : MOBILE
       Effective Dates        : 1984-01-01/1997-12-31
       Notes                  :

7.4  Instrumentation Type     : VLBI 7224
       Status                 : PERMANENT
       Effective Dates        : 1983-01-01/CCYY-MM-DD
       Notes                  :

7.5  Instrumentation Type     : LLR WLRS
       Status                 : PERMANENT
       Effective Dates        : 1991-01-01/CCYY-MM-DD
       Notes                  :

7.6  Instrumentation Type     : TIGO VLBI
       Status                 : PERMANENT / MOBIL
       Effective Dates        : 1997-01-01/2001-12-01
       Notes                  : TIGO Transportable Integrated Geodetic
                              : Observatory

7.7  Instrumentation Type     : TIGO SLR
       Status                 : PERMANENT / MOBIL
       Effective Dates        : 1997-01-01/2001-12-01
       Notes                  : TIGO Transportable Integrated Geodetic
                              : Observatory

7.8  Instrumentation Type     : PRARE
       Status                 : PERMANENT
       Effective Dates        : 1994-01-01/CCYY-MM-DD
       Notes                  :

7.9  Instrumentation Type     : Super Conducting Gravimeter
       Status                 : PERMANENT
       Effective Dates        : 1989-01-01/CCYY-MM-DD
       Notes                  :

7.10  Instrumentation Type    : Very Broad Band Seismograph
       Status                 : PERMANENT
       Effective Dates        : 1991-01-01/CCYY-MM-DD
       Notes                  :

7.11  Instrumentation Type    : Transit NNSS-DMA Doppler
       Status                 : PERMANENT
       Effective Dates        : 1984-01-01/1993-12-31
       Notes                  :

7.12  Instrumentation Type    : TI-4100 GPS
       Status                 : PERMANENT
       Effective Dates        : 1987-11-12/1989-07-27
       Notes                  : CIGNET GPS Station

7.13  Instrumentation Type    : MiniMac 2816 AT GPS
       Status                 : PERMANENT
       Effective Dates        : 1989-05-23/1991-07-23
       Notes                  : CIGNET GPS Station

7.14  Instrumentation Type    : ROGUE SNR-800 GPS WETT # 1200
       Status                 : PERMANENT
       Effective Dates        : 1991-07-23/1997-02-02
       Notes                  : CIGNET / CORE / IGS Station

7.15 Instrumentation Type     : TRIMBLE 4000SSI WTZT # 1201
       Status                 : PERMANENT
       Effective Dates        : 1997-02-06/2005-05-17
       Notes                  :

7.16 Instrumentation Type     : ROGUE SNR-8000 GPS WTZR # 1202
       Status                 : PERMANENT
       Effective Dates        : 1995-02-09/1999-06-17
       Notes                  : IGS Referece Frame Station

7.17 Instrumentation Type     : AOA SNR-8000 ACT GPS WTZR # 1202
       Status                 : PERMANENT
       Effective Dates        : 1999-06-17/2005-05-04
       Notes                  : IGS Reference Frame Station

7.18 Instrumentation Type     : TPS E_GGD GPS+GLONASS WTZR # 1202
       Status                 : PERMANENT
       Effective Dates        : 2005-05-04/2007-05-29
       Notes                  : IGS Reference Frame Station

7.19 Instrumentation Type     : TPS NETG3 GPS+GLONASS WTZR # 1202
       Status                 : PERMANENT
       Effective Dates        : 2007-05-29/2008-01-25
       Notes                  : IGS Reference Frame Station

7.20 Instrumentation Type     : LEICA GRX1200GGPRO GPS+GLONASS WTZR # 1202
       Status                 : PERMANENT
       Effective Dates        : 2008-01-25/CCYY-MM-DD
       Notes                  : IGS Reference Frame Station

7.21 Instrumentation Type     : 3SNAV R101 OLD GPS+GLONASS WTZG # 1203
       Status                 : PERMANENT
       Effective Dates        : 1998-10-14/1999-09-04
       Notes                  :

7.22 Instrumentation Type     : ASHTECH Z-XII3 GPS WTZA # 1204
       Status                 : PERMANENT
       Effective Dates        : 1997-11-25/2001-11-08
       Notes                  :

7.23 Instrumentation Type     : ASHTECH Z-XII3T GPS WTZA # 1204
       Status                 : PERMANENT
       Effective Dates        : 2001-11-08/CCYY-MM-DD
       Notes                  :

7.24 Instrumentation Type     : ASHTECH Z-18 GPS+GLONASS WTZZ #1205
       Status                 : PERMANENT
       Effective Dates        : 1999-02-01/2003-06-10
       Notes                  :

7.25 Instrumentation Type     : JPS LEGACY GPS+GLONASS WTZZ #1205
       Status                 : PERMANENT
       Effective Dates        : 2003-06-10/2003-11-06
       Notes                  :

7.26 Instrumentation Type     : JPS E_GGD GPS+GLONASS WTZZ #1205
       Status                 : PERMANENT
       Effective Dates        : 2003-11-06/2005-05-04
       Notes                  :

7.27 Instrumentation Type     : TPS E_GGD GPS+GLONASS WTZZ #1205
       Status                 : PERMANENT
       Effective Dates        : 2005-05-04/CCYY-MM-DD
       Notes                  :

7.28 Instrumentation Type     : LEICA GRX1200PRO GPS WTZL #1206
       Status                 : PERMANENT
       Effective Dates        : 2004-03-23/2007-09-05
       Notes                  :

7.29 Instrumentation Type     : LEICA GRX1200GGPRO GPS+GLONASS WTZL #1206
       Status                 : PERMANENT
       Effective Dates        : 2007-09-05/2008-01-28
       Notes                  :

7.30 Instrumentation Type     : TPS NETG3 GPS+GLONASS WTZL #1206
       Status                 : PERMANENT
       Effective Dates        : 2008-01-28/CCYY-MM-DD
       Notes                  :

7.31 Instrumentation Type     : SEPT POLARX2 GPS WTZS
       Status                 : PERMANENT
       Effective Dates        : 2005-07-27/CCYY-MM-DD
       Notes

7.32 Instrumentation Type     : Ringlaser Gyroscope
       Status                 : PERMANENT
       Effective Dates        : 2001-09-01/CCYY-MM-DD
       Notes                  :

7.x  Instrumentation Type     : (GPS/GLONASS/DORIS/PRARE/SLR/VLBI/TIME/etc)
       Status                 : (PERMANENT/MOBILE)
       Effective Dates        : (CCYY-MM-DD/CCYY-MM-DD)
       Notes                  : (multiple lines)

8.   Meteorological Instrumentation

8.1.1  Humidity Sensor Model  : 809
       Manufacturer           : Lamprecht, Goettingen
       Serial Number          :
       Data Sampling Interval : 900
       Accuracy (% rel h)     : 1.5
       Aspiration             : PERNIX element
       Height Diff to Ant     : 0
       Calibration date       : (CCYY-MM-DD)
       Effective Dates        : 2001-07-03/2002-05-28
       Notes                  : Kindly use the humidity data
                              : from the WTZR GPS Permanent
                              : Tracking Station

8.1.2  Humidity Sensor Model  : HMP45A-P
       Manufacturer           : Vaisala Oyi
       Serial Number          : W5040013
       Data Sampling Interval : 900
       Accuracy (% rel h)     : 1.0
       Aspiration             : HUMICAP 180 CAPACITIVY THIN FILM POLYMER
       Height Diff to Ant     : -0.5
       Calibration date       : (CCYY-MM-DD)
       Effective Dates        : 2002-05-28/CCYY-MM-DD
       Notes                  : New Humidity Sensor for WTZJ
                              : is 0.5 m above the antenna

8.1.x  Humidity Sensor Model  :
       Manufacturer           :
       Serial Number          :
       Data Sampling Interval : (sec)
       Accuracy (% rel h)     : (% rel h)
       Aspiration             : (UNASPIRATED/NATURAL/FAN/etc)
       Height Diff to Ant     : (m)
       Calibration date       : (CCYY-MM-DD)
       Effective Dates        : (CCYY-MM-DD/CCYY-MM-DD)
       Notes                  : (multiple lines)

8.2.1  Pressure Sensor Model  : DIGIQUARTZ MODEL 740
       Manufacturer           : Parascientific Inc.
       Serial Number          :
       Data Sampling Interval : 900
       Accuracy               : 0.1
       Height Diff to Ant     : 10.5
       Calibration date       : (CCYY-MM-DD)
       Effective Dates        : 2001-07-03/2002-05-28
       Notes                  : Kindly use the pressure data
                              : from the WTZR GPS Permanent
                              : Tracking Station. This pressure
                              : Sensor is 10.5 m below the
                              : antenna.

8.2.2  Pressure Sensor Model  : PTU200CAA1AB
       Manufacturer           : Vaisala Oyi
       Serial Number          : W5110013
       Data Sampling Interval : 900
       Accuracy               : 0.1
       Height Diff to Ant     : -0.5
       Calibration date       : (CCYY-MM-DD)
       Effective Dates        : 2002-05-28/CCYY-MM-DD
       Notes                  : New Pressure Sensor for WTZJ
                              : is 0.5 m above the antenna

8.2.x  Pressure Sensor Model  :
       Manufacturer           :
       Serial Number          :
       Data Sampling Interval : (sec)
       Accuracy               : (hPa)
       Height Diff to Ant     : (m)
       Calibration date       : (CCYY-MM-DD)
       Effective Dates        : (CCYY-MM-DD/CCYY-MM-DD)
       Notes                  : (multiple lines)

8.3.1  Temp. Sensor Model     : 809
       Manufacturer           : Lamprecht, Goettingen
       Serial Number          :
       Data Sampling Interval : 900
       Accuracy               : 0.3
       Aspiration             : PLATINUM PROBE
       Height Diff to Ant     : 0
       Calibration date       : (CCYY-MM-DD)
       Effective Dates        : 2001-07-03/2002-05-28
       Notes                  : Kindly use the temperature data
                              : from the WTZR GPS Permanent
                              : Tracking Station.

8.3.2  Temp. Sensor Model     : HMP45A-P
       Manufacturer           : Vaisala Oyi
       Serial Number          : W5040013
       Data Sampling Interval : 900
       Accuracy               : 0.2
       Aspiration             : RESISTIVE PLATINUM PROBE
       Height Diff to Ant     : -0.5
       Calibration date       : (CCYY-MM-DD)
       Effective Dates        : 2002-05-28/CCYY-MM-DD
       Notes                  : New Temperature Sensor for
                              : WTZJ is 0.5 m above the antenna

8.3.x  Temp. Sensor Model     :
       Manufacturer           :
       Serial Number          :
       Data Sampling Interval : (sec)
       Accuracy               : (hPa)
       Aspiration             : (UNASPIRATED/NATURAL/FAN/etc)
       Height Diff to Ant     : (m)
       Calibration date       : (CCYY-MM-DD)
       Effective Dates        : (CCYY-MM-DD/CCYY-MM-DD)
       Notes                  : (multiple lines)

8.4.1  Water Vapor Radiometer : CTGR129502
       Manufacturer           : Captec, Bern / ETH, Zuerich
       Serial Number          :
       Distance to Antenna    : 27
       Height Diff to Ant     : 1
       Calibration date       : (CCYY-MM-DD)
       Effective Dates        : 1997-04-01/CCYY-MM-DD
       Notes                  :

8.4.x  Water Vapor Radiometer :
       Manufacturer           :
       Serial Number          :
       Distance to Antenna    : (m)
       Height Diff to Ant     : (m)
       Calibration date       : (CCYY-MM-DD)
       Effective Dates        : (CCYY-MM-DD/CCYY-MM-DD)
       Notes                  : (multiple lines)

8.5.x  Other Instrumentation  :

9.   Local Ongoing Conditions Possibly Affecting Computed Position

9.1.x  Radio Interferences    : (TV/CELL PHONE ANTENNA/RADAR/etc)
       Observed Degredations  : (SN RATIO/DATA GAPS/etc)
       Effective Dates        : (CCYY-MM-DD/CCYY-MM-DD)
       Additional Information : (multiple lines)

9.2.x  Multipath Sources      : METAL ROOF/DOME/VLBI ANTENNA/etc)
       Effective Dates        : (CCYY-MM-DD/CCYY-MM-DD)
       Additional Information : (multiple lines)

9.3.x  Signal Obstructions    : (TREES/BUILDLINGS/etc)
       Effective Dates        : (CCYY-MM-DD/CCYY-MM-DD)
       Additional Information : (multiple lines)

10.  Local Episodic Effects Possibly Affecting Data Quality

10.x Date                     : (CCYY-MM-DD/CCYY-MM-DD)
     Event                    : (TREE CLEARING/CONSTRUCTION/etc)

11.  On-Site, Point of Contact Agency Information

     Agency                   : Bundesamt fuer Kartographie
                              : und Geodaesie
                              : Geodetical Observatory Wettzell
     Preferred Abbreviation   : BKG
     Mailing Address          : Bundesamt fuer Kartographie
                              : und Geodaesie
                              : Geodetical Observatory Wettzell
                              : Sackenrieder Strasse 25
                              : D-93444 Bad Koetzting
                              : Germany
     Primary Contact
       Contact Name           : Dr. Reiner Dassing
       Telephone (primary)    : ++49-9941-603112
       Telephone (secondary)  : ++49-9941-6030
       Fax                    : ++49-9941-603222
       E-mail                 : [email-masked]
     Secondary Contact
       Contact Name           : Uwe Hessels
       Telephone (primary)    : ++49-9941-603208
       Telephone (secondary)  : ++49-9941-6030
       Fax                    : ++49-9941-603222
       E-mail                 : [email-masked]
     Additional Information   : Klaus Roettcher
                              : ++49-9941-603115
                              : [email-masked]
                              : Rudi Stoeger
                              : ++49-9941-603117
                              : [email-masked]

12.  Responsible Agency (if different from 11.)

     Agency                   : Bundesamt fuer Kartographie
                              : und Geodaesie
     Preferred Abbreviation   : BKG
     Mailing Address          : Bundesamt fuer Kartographie
                              : und Geodaesie
                              : Department Leizig
                              : Karl-Rothe-Strasse 10-14
                              : D-04105 Leipzig
                              : Germany
     Primary Contact
       Contact Name           : Matthias Groeschel
       Telephone (primary)    : ++49-341-5643-432
       Telephone (secondary)  : ++49-341-5643-0
       Fax                    : ++49-341-5643-415
       E-mail                 : [email-masked]
     Secondary Contact
       Contact Name           : Peter Franke (in Frankfurt)
       Telephone (primary)    : ++49-69-6333-279
       Telephone (secondary)  : ++49-69-6333-1
       Fax                    : ++49-69-6333-425
       E-mail                 : [email-masked]
     Additional Information   :

13.  More Information

     Primary Data Center      : BKGI
     Secondary Data Center    : CDDIS
     URL for More Information : www.bkg.bund.de.de  www.fs.wettzell.de
       Hardcopy on File
       Site Map               : (Y or URL)
       Site Diagram           : Y
       Horizon Mask           : Y
       Monument Description   : Y
       Site Pictures          : Y
     Additional Information   : (multiple lines)

     Antenna Graphics with Dimensions

TRM29659.00
>From 2001-07-03 until 2002-04-11

                         -----
                     /     +     \                        <--  0.128  L2
                    |      +      |                       <--  0.110  L1
  +--------------------------------------------------+    <--  0.102  TCR
  |                                                  |
  |                                                  |
  |                                                  |
  |                                                  |
+-+--------------------------------------------------+-+  <--  0.038
+-------------------+-------------+--------------------+  <--  0.035  BCR
                    |             |
                   =|             |
                    +------x------+                       <--  0.000  BPA=ARP

<--                      0.381                       -->

JPSREGANT_SD_E
>From 2002-04-11 until 2005-08-02

                +-------------------------+ <--------------Plastic Housing for
               /    -------------------    \               Antenna/Choke Ring
              /    /         +         \    \                <-- 0.1196 L2
   +---------+     |         +         |     +---------+     <-- 0.1060 L1
  | +-------------------------------------------------+ |    <-- 0.xxxx TCR
 /  |                                                 |  \
|   |                                                 |   |
|   |                                                 |   |
|   |                                                 |   |
 \+-----------------------------------------------------+/
 |+-----------------------------------------------------+|   <-- 0.xxxx BCR
  \______________  |                   |  ______________/
                 \ |                   | /
                  \+---------x---------+/                    <-- 0.0000 BPA=ARP

TRM29659.00
>From 2005-08-02 until 2009-12-16

                         -----
                     /     +     \                        <--  0.128  L2
                    |      +      |                       <--  0.110  L1
  +--------------------------------------------------+    <--  0.102  TCR
  |                                                  |
  |                                                  |
  |                                                  |
  |                                                  |
+-+--------------------------------------------------+-+  <--  0.038
+-------------------+-------------+--------------------+  <--  0.035  BCR
                    |             |
                   =|             |
                    +------x------+                       <--  0.000  BPA=ARP

<--                      0.381                       -->

LEIAR25
>From 2009-12-16

                         +-------+                          <-- 0.1995
                       /     +     \
                      |      +      |
                  |---+-------------+----|		    <-- 0.1614 TCR
              |------------------------------|
          |--------------------------------------|
      |----------------------------------------------|
    +--------------------------------------------------+
    |                                                  |
    |                                                  |
    |                                                  |
  +-+--------------------------------------------------+-+
  +-------------------+-------------+--------------------+  <-- 0.0320 BCR
                      |             |
                      |             |
                      +------x------+                       <-- 0.0000 BPA=ARP

  <--                      0.3802                      -->

   ARP: Antenna Reference Point
   L1 : L1 Phase Center                   L2 : L2 Phase Center
   TCR: Top of Chokering                  BCR: Bottom of Chokering
   TGP: Top of Ground Plane               BGP: Bottom of Ground Plane
   TPA: Top of Preamplifier               BPA: Bottom of Preamplifier
   TOP: Top of Pole

All dimensions are in meters.
     International GPS Service
     WTZJ Site Information Form
     See Instructions at:
       ftp://igscb.jpl.nasa.gov/pub/station/general/sitelog_instr.txt

0.   Form

     Prepared by (full name)  : Rudolf Stoeger
                              : Peter Franke
                              : Matthias Groeschel
     Date Prepared            : 2009-12-17
     Report Type              : UPDATE
     If Update:
      Previous Site Log       : wtzj_20091216.log
      Modified/Added Sections : 4.4, 4.5

1.   Site Identification of the GNSS Monument

     Site Name                : Wettzell / Germany
     Four Character ID        : WTZJ
     Monument Inscription     : Pillar 1203
     IERS DOMES Number        : 14201M012
     CDP Number               :
     Monument Description     : STEEL PLATE ON CONCRETE SURVEY TOWER
       Height of the Monument : 0.8
       Monument Foundation    : SCREW ANCHORED TO CONCRETE SURVEY TOWER
       Foundation Depth       :
     Marker Description       :
     Date Installed           : 2001-07-03T00:00Z
     Geologic Characteristic  : Bedrock
       Bedrock Type           : Cordirit-Silimanit-Gneiss
       Bedrock Condition      : Weathered
       Fracture Spacing       : 50-200 cm and over 200 cm
       Fault zones nearby     : NO
         Distance/activity    :
       Additional Information : The antenna is 7.5 m above
                              : the ground.
                              : On the same pillar 1203,
                              : DOMES No. 14201M012, also
                              : a 3SNAV R101 GPS+GLONASS Receiver,
                              : Four Character ID WTZG, was
                              : observing from 14-OCT-1998 until
                              : 04-SEP-1999.
                              : In addition and parallel a
                              : LEICA GRX1200GGPRO GPS+GLONASS Receiver,
                              : Four Character ID WTZR, IGS
                              : Reference Frame Station, on
                              : pillar 1202, Domes No. 14201M010,
                              : an ASHTECH Z-XII3T Metronome GPS
                              : Receiver, Four character ID WTZA,
                              : on pillar 1204, Domes No. 14201M013,
                              : a TPS E_GGD GPS+GLONASS Receiver,
                              : Four character ID WTZZ, on pillar
                              : 1205, Domes No. 14201M014, a 
                              : TPS NETG3 GPS+GLONASS Receiver,
                              : Four character ID WTZL, on
                              : pillar 1206, DOMES No. 14201M022
                              : and a SEPT POLARX2, Four Character
                              : ID WTZS, DOMES No. 14201M015
                              : are observing.
                              : Prior we observed here in the
                              : Fundamentalstation Wettzell with a
                              : TI-4100, a MiniMac 2816 AT, a
                              : Rogue SNR-800, our # 1200, Four Character
                              : ID WETT, Domes No. 14201M009,
                              : the 3SNAV R101 GPS+GLONASS
                              : Receiver, Four character ID WTZG,
                              : on pillar 1203, Domes No.
                              : 14201M012, and a TRIMBLE 4000 SSI,
                              : Four character ID WTZT.
                              : Further informations and effective
                              : dates see sections 5 and 7.

2.   Site Location Information

     City or Town             : Bad Koetzting
     State or Province        : Bavaria
     Country                  : Germany
     Tectonic Plate           : Eurasian
     Approximate Position
       X coordinate (m)       : 4075580.0194
       Y coordinate (m)       : 931855.2287
       Z coordinate (m)       : 4801568.1824
       Latitude (N is +)      : +490839.12
       Longitude (E is +)     : +0125244.15
       Elevation (m,ellips.)  : 00665.9
     Additional Information   :

3.   GNSS Receiver Information

3.1  Receiver Type            : JPS LEGACY
     Satellite System         : GPS+GLONASS
     Serial Number            : MT312211422
     Firmware Version         : EURO_3/2.1 JAN,03
     Elevation Cutoff Setting : 0
     Date Installed           : 2001-07-03T12:00Z
     Date Removed             : 2002-10-09T12:00Z
     Tempearture Stabiliz.    : 2.0
     Additional Information   :

3.2  Receiver Type            : JPS LEGACY
     Satellite System         : GPS+GLONASS
     Serial Number            : MT312211422
     Firmware Version         : EURO_3/2.2p2
     Elevation Cutoff Setting : 0
     Date Installed           : 2002-10-09T12:00Z
     Date Removed             : 2003-11-24T14:00Z
     Tempearture Stabiliz.    : 2.0
     Additional Information   :

3.3  Receiver Type            : JPS LEGACY
     Satellite System         : GPS+GLONASS
     Serial Number            : MT312211422
     Firmware Version         : EURO_3/2.3p3
     Elevation Cutoff Setting : 0
     Date Installed           : 2003-11-24T14:00Z
     Date Removed             : 2004-05-14T12:00Z
     Tempearture Stabiliz.    : 2.0
     Additional Information   :

3.4  Receiver Type            : JPS LEGACY
     Satellite System         : GPS+GLONASS
     Serial Number            : MT312211422
     Firmware Version         : Euro_3/2.3p4
     Elevation Cutoff Setting : 0
     Date Installed           : 2004-05-14T12:00Z
     Date Removed             : 2007-10-30T11:30Z
     Tempearture Stabiliz.    : 2.0
     Additional Information   : Additional Number
                              : LE 2808 on the
                              : type level

3.5  Receiver Type            : JPS LEGACY
     Satellite System         : GPS+GLONASS
     Serial Number            : MT312211422
     Firmware Version         : 2.6.0 OCT,24,2007 O
     Elevation Cutoff Setting : 0
     Date Installed           : 2007-10-30T11:30Z
     Date Removed             : 2008-01-24T08:44Z
     Tempearture Stabiliz.    : 2.0
     Additional Information   : Additional Number
                              : LE 2808 on the
                              : type level

3.6  Receiver Type            : JPS LEGACY
     Satellite System         : GPS+GLONASS
     Serial Number            : MT312211422
     Firmware Version         : 2.6.1 JAN,10,2008
     Elevation Cutoff Setting : 0
     Date Installed           : 2008-01-24T08:45Z
     Date Removed             : CCYY-MM-DDThh:mmZ
     Tempearture Stabiliz.    : 2.0
     Additional Information   : Additional Number
                              : LE 2808 on the
                              : type level

4.   GNSS Antenna Information

4.1  Antenna Type             : TRM29659.00     NONE
     Serial Number            : 0220180414
     Antenna Reference Point  : BPA
     Marker->ARP Up Ecc. (m)  : 000.0760
     Marker->ARP North Ecc(m) : 000.0000
     Marker->ARP East Ecc(m)  : 000.0000
     Alignment from True N    : 0
     Antenna Radome Type      : NONE
     Radome Serial Number     :
     Antenna Cable Type       : RG214
     Antenna Cable Length     : 30
     Date Installed           : 2001-07-03T12:00Z
     Date Removed             : 2002-04-11T09:30Z
     Additional Information   :

4.2  Antenna Type             : JPSREGANT_SD_E  NONE
     Serial Number            : RA0295
     Antenna Reference Point  : BPA
     Marker->ARP Up Ecc. (m)  : 000.0740
     Marker->ARP North Ecc(m) : 000.0000
     Marker->ARP East Ecc(m)  : 000.0000
     Alignment from True N    : 0
     Antenna Radome Type      : NONE
     Radome Serial Number     :
     Antenna Cable Type       : RG214
     Antenna Cable Length     : 30
     Date Installed           : 2002-04-11T09:30Z
     Date Removed             : 2005-08-02T09:00Z
     Additional Information   :

4.3  Antenna Type             : TRM29659.00     NONE
     Serial Number            : 0220175208
     Antenna Reference Point  : BPA
     Marker->ARP Up Ecc. (m)  : 000.0740
     Marker->ARP North Ecc(m) : 000.0000
     Marker->ARP East Ecc(m)  : 000.0000
     Alignment from True N    : 0
     Antenna Radome Type      : NONE
     Radome Serial Number     :
     Antenna Cable Type       : RG214
     Antenna Cable Length     : 30
     Date Installed           : 2005-08-02T09:00Z
     Date Removed             : 2009-12-16T11:59Z
     Additional Information   : Antenna calibrated absolute
                              : by GEO++ GmbH Garbsen,
                              : 2000-10-31.

4.4  Antenna Type             : LEIAR25         NONE
     Serial Number            : 08430008
     Antenna Reference Point  : BPA
     Marker->ARP Up Ecc. (m)  : 000.0740
     Marker->ARP North Ecc(m) : 000.0000
     Marker->ARP East Ecc(m)  : 000.0000
     Alignment from True N    : 0
     Antenna Radome Type      : NONE
     Radome Serial Number     :
     Antenna Cable Type       : RG214
     Antenna Cable Length     : 30
     Date Installed           : 2009-12-16T12:00Z
     Date Removed             : 2009-12-17T07:35Z
     Additional Information   : Antenna calibrated absolute
                              : by GEO++ GmbH Garbsen,
                              : 2008-11-20 together with
                              : radome LEIT.
                              : The radome LEIT is not installed.

4.5  Antenna Type             : LEIAR25         LEIT
     Serial Number            : 08430008
     Antenna Reference Point  : BPA
     Marker->ARP Up Ecc. (m)  : 000.0740
     Marker->ARP North Ecc(m) : 000.0000
     Marker->ARP East Ecc(m)  : 000.0000
     Alignment from True N    : 0
     Antenna Radome Type      : LEIT
     Radome Serial Number     :
     Antenna Cable Type       : RG214
     Antenna Cable Length     : 30
     Date Installed           : 2009-12-17T07:35Z
     Date Removed             : (CCYY-MM-DDThh:mmZ)
     Additional Information   : Antenna calibrated absolute
                              : by GEO++ GmbH Garbsen,
                              : 2008-11-20 together with
                              : radome LEIT.

4.x  Antenna Type             : (A20, from rcvr_ant.tab; see instructions)
     Serial Number            : (A*, but note the first A5 is used in SINEX)
     Antenna Reference Point  : (BPA/BCR/XXX from "antenna.gra"; see instr.)
     Marker->ARP Up Ecc. (m)  : (F8.4)
     Marker->ARP North Ecc(m) : (F8.4)
     Marker->ARP East Ecc(m)  : (F8.4)
     Alignment from True N    : (deg; + is clockwise/east)
     Antenna Radome Type      : (A4 from rcvr_ant.tab; see instructions)
     Radome Serial Number     :
     Antenna Cable Type       : (vendor & type number)
     Antenna Cable Length     : (m)
     Date Installed           : (CCYY-MM-DDThh:mmZ)
     Date Removed             : (CCYY-MM-DDThh:mmZ)
     Additional Information   : (multiple lines)

5.  Surveyed Local Ties

5.1  Tied Marker Name         : VLBI Ref Point
     Tied Marker Usage        : VLBI
     Tied Marker CDP Number   : 7224
     Tied Marker DOMES Number : 14201S004
     Differential Components from GNSS Marker to the tied monument (ITRS)
       dx (m)                 : -40.1314
       dy (m)                 : -119.9655
       dz (m)                 : +61.1925
     Accuracy (mm)            : 1.3
     Survey method            : TRIANGULATION
     Date Measured            : 2004-10-01T10:00Z
     Additional Information   :

5.2  Tied Marker Name         : SLR WLRS
     Tied Marker Usage        : SLR
     Tied Marker CDP Number   : 8834
     Tied Marker DOMES Number : 14201S018
     Differential Components from GNSS Marker to the tied monument (ITRS)
       dx (m)                 : -3.1580
       dy (m)                 : -69.7708
       dz (m)                 : +15.3895
     Accuracy (mm)            : 0.6
     Survey method            : TRIANGULATION
     Date Measured            : 2004-10-01T10:00Z
     Additional Information   :

5.3  Tied Marker Name         : TIGO VLBI
     Tied Marker Usage        : VLBI
     Tied Marker CDP Number   : 759324012
     Tied Marker DOMES Number : 14201S100
     Differential Components from GNSS Marker to the tied monument (ITRS)
       dx (m)                 : -8.2314
       dy (m)                 : -100.3115
       dz (m)                 : +15.1224
     Accuracy (mm)            : 0.9
     Survey method            : TRIANGULATION
     Date Measured            : 2004-10-01T10:00Z
     Additional Information   :

5.4  Tied Marker Name         : TIGO SLR
     Tied Marker Usage        : SLR
     Tied Marker CDP Number   : 75947901
     Tied Marker DOMES Number : 14201M200
     Differential Components from GNSS Marker to the tied monument (ITRS)
       dx (m)                 : +15.0227
       dy (m)                 : -122.2774
       dz (m)                 : -0.6106
     Accuracy (mm)            : 1.5
     Survey method            : TRIANGULATION
     Date Measured            : 2004-10-01T10:00Z
     Additional Information   :

5.5  Tied Marker Name         : 1200 ROGUE SNR-800 = WETT
     Tied Marker Usage        : GPS
     Tied Marker CDP Number   :
     Tied Marker DOMES Number : 14201M009
     Differential Components from GNSS Marker to the tied monument (ITRS)
       dx (m)                 : -1.4392
       dy (m)                 : -2.5501
       dz (m)                 : +1.8658
     Accuracy (mm)            : 1
     Survey method            : TRIANGULATION
     Date Measured            : 1995-02-13T15:00Z
     Additional Information   :

5.6  Tied Marker Name         : 1202 LEICA GRX1200GGPRO = WTZR
     Tied Marker Usage        : GPS+GLONASS
     Tied Marker CDP Number   :
     Tied Marker DOMES Number : 14201M010
     Differential Components from GNSS Marker to the tied monument (ITRS)
       dx (m)                 : +0.6658
       dy (m)                 : -1.5691
       dz (m)                 : -0.1282
     Accuracy (mm)            : 0.4
     Survey method            : TRIANGULATION
     Date Measured            : 2004-10-01T10:00Z
     Additional Information   : IGS Reference Frame Station

5.7  Tied Marker Name         : 1201 TRIMBLE 4000SSI = WTZT
     Tied Marker Usage        : GPS
     Tied Marker CDP Number   :
     Tied Marker DOMES Number : 14201M011
     Differential Components from GNSS Marker to the tied monument (ITRS)
       dx (m)                 : -2.4054
       dy (m)                 : +0.0527
       dz (m)                 : +2.0291
     Accuracy (mm)            : 0.4
     Survey method            : TRIANGULATION
     Date Measured            : 2004-10-01T10:00Z
     Additional Information   :

5.8  Tied Marker Name         : 1204 ASHTECH Z-XII3T = WTZA
     Tied Marker Usage        : GPS
     Tied Marker CDP Number   :
     Tied Marker DOMES Number : 14201M013
     Differential Components from GNSS Marker to the tied monument (ITRS)
       dx (m)                 : -1.5250
       dy (m)                 : -2.5706
       dz (m)                 : +1.7634
     Accuracy (mm)            : 0.4
     Survey method            : TRIANGULATION
     Date Measured            : 2004-10-01T10:00Z
     Additional Information   :

5.9  Tied Marker Name         : 1205 TPS E_GGD = WTZZ
     Tied Marker Usage        : GPS+GLONASS
     Tied Marker CDP Number   :
     Tied Marker DOMES Number : 14201M014
     Differential Components from GNSS Marker to the tied monument (ITRS)
       dx (m)                 : -0.4575
       dy (m)                 : -2.2556
       dz (m)                 : +0.7776
     Accuracy (mm)            : 0.4
     Survey method            : TRIANGULATION
     Date Measured            : 2004-10-01T10:00Z
     Additional Information   :

5.10 Tied Marker Name         : 1206 LEICA GRX1200PRO = WTZL
     Tied Marker Usage        : GPS
     Tied Marker CDP Number   :
     Tied Marker DOMES Number : 14201M022
     Differential Components from GNSS Marker to the tied monument (ITRS)
       dx (m)                 : -1.9769
       dy (m)                 : -1.2622
       dz (m)                 : +1.8975
     Accuracy (mm)            : 0.4
     Survey method            : TRIANGULATION
     Date Measured            : 2004-10-01T10:00Z
     Additional Information   :

5.11 Tied Marker Name         : SEPT POLARX2 = WTZS
     Tied Marker Usage        : GPS
     Tied Marker CDP Number   :
     Tied Marker DOMES Number : 14201M015
     Differential Components from GNSS Marker to the tied monument (ITRS)
       dx (m)                 :
       dy (m)                 :
       dz (m)                 :
     Accuracy (mm)            :
     Survey method            :
     Date Measured            :
     Additional Information   :

5.x  Tied Marker Name         :
     Tied Marker Usage        : (SLR/VLBI/LOCAL CONTROL/FOOTPRINT/etc)
     Tied Marker CDP Number   : (A4)
     Tied Marker DOMES Number : (A9)
     Differential Components from GNSS Marker to the tied monument (ITRS)
       dx (m)                 :
       dy (m)                 :
       dz (m)                 :
     Accuracy (mm)            : (mm)
     Survey method            : (GPS CAMPAIGN/TRILATERATION/TRIANGULATION/etc)
     Date Measured            : (CCYY-MM-DDThh:mmZ)
     Additional Information   : (multiple lines)

6.   Frequency Standard

6.1  Standard Type            : INTERNAL
       Input Frequency        :
       Effective Dates        : 2001-07-03/CCYY-MM-DD
       Notes                  :

6.x  Standard Type            : (INTERNAL or EXTERNAL H-MASER/CESIUM/etc)
       Input Frequency        : (if external)
       Effective Dates        : (CCYY-MM-DD/CCYY-MM-DD)
       Notes                  : (multiple lines)

7.   Collocation Information

7.1  Instrumentation Type     : SLR WLRS 8834
       Status                 : PERMANENT
       Effective Dates        : 1988-01-01/CCYY-MM-DD
       Notes                  :

7.2  Instrumentation Type     : SLR 7834
       Status                 : PERMANENT
       Effective Dates        : 1977-01-01/1991-12-31
       Notes                  :

7.3  Instrumentation Type     : SLR MTLRS 1
       Status                 : MOBILE
       Effective Dates        : 1984-01-01/1997-12-31
       Notes                  :

7.4  Instrumentation Type     : VLBI 7224
       Status                 : PERMANENT
       Effective Dates        : 1983-01-01/CCYY-MM-DD
       Notes                  :

7.5  Instrumentation Type     : LLR WLRS
       Status                 : PERMANENT
       Effective Dates        : 1991-01-01/CCYY-MM-DD
       Notes                  :

7.6  Instrumentation Type     : TIGO VLBI
       Status                 : PERMANENT / MOBIL
       Effective Dates        : 1997-01-01/2001-12-01
       Notes                  : TIGO Transportable Integrated Geodetic
                              : Observatory

7.7  Instrumentation Type     : TIGO SLR
       Status                 : PERMANENT / MOBIL
       Effective Dates        : 1997-01-01/2001-12-01
       Notes                  : TIGO Transportable Integrated Geodetic
                              : Observatory

7.8  Instrumentation Type     : PRARE
       Status                 : PERMANENT
       Effective Dates        : 1994-01-01/CCYY-MM-DD
       Notes                  :

7.9  Instrumentation Type     : Super Conducting Gravimeter
       Status                 : PERMANENT
       Effective Dates        : 1989-01-01/CCYY-MM-DD
       Notes                  :

7.10  Instrumentation Type    : Very Broad Band Seismograph
       Status                 : PERMANENT
       Effective Dates        : 1991-01-01/CCYY-MM-DD
       Notes                  :

7.11  Instrumentation Type    : Transit NNSS-DMA Doppler
       Status                 : PERMANENT
       Effective Dates        : 1984-01-01/1993-12-31
       Notes                  :

7.12  Instrumentation Type    : TI-4100 GPS
       Status                 : PERMANENT
       Effective Dates        : 1987-11-12/1989-07-27
       Notes                  : CIGNET GPS Station

7.13  Instrumentation Type    : MiniMac 2816 AT GPS
       Status                 : PERMANENT
       Effective Dates        : 1989-05-23/1991-07-23
       Notes                  : CIGNET GPS Station

7.14  Instrumentation Type    : ROGUE SNR-800 GPS WETT # 1200
       Status                 : PERMANENT
       Effective Dates        : 1991-07-23/1997-02-02
       Notes                  : CIGNET / CORE / IGS Station

7.15 Instrumentation Type     : TRIMBLE 4000SSI WTZT # 1201
       Status                 : PERMANENT
       Effective Dates        : 1997-02-06/2005-05-17
       Notes                  :

7.16 Instrumentation Type     : ROGUE SNR-8000 GPS WTZR # 1202
       Status                 : PERMANENT
       Effective Dates        : 1995-02-09/1999-06-17
       Notes                  : IGS Referece Frame Station

7.17 Instrumentation Type     : AOA SNR-8000 ACT GPS WTZR # 1202
       Status                 : PERMANENT
       Effective Dates        : 1999-06-17/2005-05-04
       Notes                  : IGS Reference Frame Station

7.18 Instrumentation Type     : TPS E_GGD GPS+GLONASS WTZR # 1202
       Status                 : PERMANENT
       Effective Dates        : 2005-05-04/2007-05-29
       Notes                  : IGS Reference Frame Station

7.19 Instrumentation Type     : TPS NETG3 GPS+GLONASS WTZR # 1202
       Status                 : PERMANENT
       Effective Dates        : 2007-05-29/2008-01-25
       Notes                  : IGS Reference Frame Station

7.20 Instrumentation Type     : LEICA GRX1200GGPRO GPS+GLONASS WTZR # 1202
       Status                 : PERMANENT
       Effective Dates        : 2008-01-25/CCYY-MM-DD
       Notes                  : IGS Reference Frame Station

7.21 Instrumentation Type     : 3SNAV R101 OLD GPS+GLONASS WTZG # 1203
       Status                 : PERMANENT
       Effective Dates        : 1998-10-14/1999-09-04
       Notes                  :

7.22 Instrumentation Type     : ASHTECH Z-XII3 GPS WTZA # 1204
       Status                 : PERMANENT
       Effective Dates        : 1997-11-25/2001-11-08
       Notes                  :

7.23 Instrumentation Type     : ASHTECH Z-XII3T GPS WTZA # 1204
       Status                 : PERMANENT
       Effective Dates        : 2001-11-08/CCYY-MM-DD
       Notes                  :

7.24 Instrumentation Type     : ASHTECH Z-18 GPS+GLONASS WTZZ #1205
       Status                 : PERMANENT
       Effective Dates        : 1999-02-01/2003-06-10
       Notes                  :

7.25 Instrumentation Type     : JPS LEGACY GPS+GLONASS WTZZ #1205
       Status                 : PERMANENT
       Effective Dates        : 2003-06-10/2003-11-06
       Notes                  :

7.26 Instrumentation Type     : JPS E_GGD GPS+GLONASS WTZZ #1205
       Status                 : PERMANENT
       Effective Dates        : 2003-11-06/2005-05-04
       Notes                  :

7.27 Instrumentation Type     : TPS E_GGD GPS+GLONASS WTZZ #1205
       Status                 : PERMANENT
       Effective Dates        : 2005-05-04/CCYY-MM-DD
       Notes                  :

7.28 Instrumentation Type     : LEICA GRX1200PRO GPS WTZL #1206
       Status                 : PERMANENT
       Effective Dates        : 2004-03-23/2007-09-05
       Notes                  :

7.29 Instrumentation Type     : LEICA GRX1200GGPRO GPS+GLONASS WTZL #1206
       Status                 : PERMANENT
       Effective Dates        : 2007-09-05/2008-01-28
       Notes                  :

7.30 Instrumentation Type     : TPS NETG3 GPS+GLONASS WTZL #1206
       Status                 : PERMANENT
       Effective Dates        : 2008-01-28/CCYY-MM-DD
       Notes                  :

7.31 Instrumentation Type     : SEPT POLARX2 GPS WTZS
       Status                 : PERMANENT
       Effective Dates        : 2005-07-27/CCYY-MM-DD
       Notes

7.32 Instrumentation Type     : Ringlaser Gyroscope
       Status                 : PERMANENT
       Effective Dates        : 2001-09-01/CCYY-MM-DD
       Notes                  :

7.x  Instrumentation Type     : (GPS/GLONASS/DORIS/PRARE/SLR/VLBI/TIME/etc)
       Status                 : (PERMANENT/MOBILE)
       Effective Dates        : (CCYY-MM-DD/CCYY-MM-DD)
       Notes                  : (multiple lines)

8.   Meteorological Instrumentation

8.1.1  Humidity Sensor Model  : 809
       Manufacturer           : Lamprecht, Goettingen
       Serial Number          :
       Data Sampling Interval : 900
       Accuracy (% rel h)     : 1.5
       Aspiration             : PERNIX element
       Height Diff to Ant     : 0
       Calibration date       : (CCYY-MM-DD)
       Effective Dates        : 2001-07-03/2002-05-28
       Notes                  : Kindly use the humidity data
                              : from the WTZR GPS Permanent
                              : Tracking Station

8.1.2  Humidity Sensor Model  : HMP45A-P
       Manufacturer           : Vaisala Oyi
       Serial Number          : W5040013
       Data Sampling Interval : 900
       Accuracy (% rel h)     : 1.0
       Aspiration             : HUMICAP 180 CAPACITIVY THIN FILM POLYMER
       Height Diff to Ant     : -0.5
       Calibration date       : (CCYY-MM-DD)
       Effective Dates        : 2002-05-28/CCYY-MM-DD
       Notes                  : New Humidity Sensor for WTZJ
                              : is 0.5 m above the antenna

8.1.x  Humidity Sensor Model  :
       Manufacturer           :
       Serial Number          :
       Data Sampling Interval : (sec)
       Accuracy (% rel h)     : (% rel h)
       Aspiration             : (UNASPIRATED/NATURAL/FAN/etc)
       Height Diff to Ant     : (m)
       Calibration date       : (CCYY-MM-DD)
       Effective Dates        : (CCYY-MM-DD/CCYY-MM-DD)
       Notes                  : (multiple lines)

8.2.1  Pressure Sensor Model  : DIGIQUARTZ MODEL 740
       Manufacturer           : Parascientific Inc.
       Serial Number          :
       Data Sampling Interval : 900
       Accuracy               : 0.1
       Height Diff to Ant     : 10.5
       Calibration date       : (CCYY-MM-DD)
       Effective Dates        : 2001-07-03/2002-05-28
       Notes                  : Kindly use the pressure data
                              : from the WTZR GPS Permanent
                              : Tracking Station. This pressure
                              : Sensor is 10.5 m below the
                              : antenna.

8.2.2  Pressure Sensor Model  : PTU200CAA1AB
       Manufacturer           : Vaisala Oyi
       Serial Number          : W5110013
       Data Sampling Interval : 900
       Accuracy               : 0.1
       Height Diff to Ant     : -0.5
       Calibration date       : (CCYY-MM-DD)
       Effective Dates        : 2002-05-28/CCYY-MM-DD
       Notes                  : New Pressure Sensor for WTZJ
                              : is 0.5 m above the antenna

8.2.x  Pressure Sensor Model  :
       Manufacturer           :
       Serial Number          :
       Data Sampling Interval : (sec)
       Accuracy               : (hPa)
       Height Diff to Ant     : (m)
       Calibration date       : (CCYY-MM-DD)
       Effective Dates        : (CCYY-MM-DD/CCYY-MM-DD)
       Notes                  : (multiple lines)

8.3.1  Temp. Sensor Model     : 809
       Manufacturer           : Lamprecht, Goettingen
       Serial Number          :
       Data Sampling Interval : 900
       Accuracy               : 0.3
       Aspiration             : PLATINUM PROBE
       Height Diff to Ant     : 0
       Calibration date       : (CCYY-MM-DD)
       Effective Dates        : 2001-07-03/2002-05-28
       Notes                  : Kindly use the temperature data
                              : from the WTZR GPS Permanent
                              : Tracking Station.

8.3.2  Temp. Sensor Model     : HMP45A-P
       Manufacturer           : Vaisala Oyi
       Serial Number          : W5040013
       Data Sampling Interval : 900
       Accuracy               : 0.2
       Aspiration             : RESISTIVE PLATINUM PROBE
       Height Diff to Ant     : -0.5
       Calibration date       : (CCYY-MM-DD)
       Effective Dates        : 2002-05-28/CCYY-MM-DD
       Notes                  : New Temperature Sensor for
                              : WTZJ is 0.5 m above the antenna

8.3.x  Temp. Sensor Model     :
       Manufacturer           :
       Serial Number          :
       Data Sampling Interval : (sec)
       Accuracy               : (hPa)
       Aspiration             : (UNASPIRATED/NATURAL/FAN/etc)
       Height Diff to Ant     : (m)
       Calibration date       : (CCYY-MM-DD)
       Effective Dates        : (CCYY-MM-DD/CCYY-MM-DD)
       Notes                  : (multiple lines)

8.4.1  Water Vapor Radiometer : CTGR129502
       Manufacturer           : Captec, Bern / ETH, Zuerich
       Serial Number          :
       Distance to Antenna    : 27
       Height Diff to Ant     : 1
       Calibration date       : (CCYY-MM-DD)
       Effective Dates        : 1997-04-01/CCYY-MM-DD
       Notes                  :

8.4.x  Water Vapor Radiometer :
       Manufacturer           :
       Serial Number          :
       Distance to Antenna    : (m)
       Height Diff to Ant     : (m)
       Calibration date       : (CCYY-MM-DD)
       Effective Dates        : (CCYY-MM-DD/CCYY-MM-DD)
       Notes                  : (multiple lines)

8.5.x  Other Instrumentation  :

9.   Local Ongoing Conditions Possibly Affecting Computed Position

9.1.x  Radio Interferences    : (TV/CELL PHONE ANTENNA/RADAR/etc)
       Observed Degredations  : (SN RATIO/DATA GAPS/etc)
       Effective Dates        : (CCYY-MM-DD/CCYY-MM-DD)
       Additional Information : (multiple lines)

9.2.x  Multipath Sources      : METAL ROOF/DOME/VLBI ANTENNA/etc)
       Effective Dates        : (CCYY-MM-DD/CCYY-MM-DD)
       Additional Information : (multiple lines)

9.3.x  Signal Obstructions    : (TREES/BUILDLINGS/etc)
       Effective Dates        : (CCYY-MM-DD/CCYY-MM-DD)
       Additional Information : (multiple lines)

10.  Local Episodic Effects Possibly Affecting Data Quality

10.x Date                     : (CCYY-MM-DD/CCYY-MM-DD)
     Event                    : (TREE CLEARING/CONSTRUCTION/etc)

11.  On-Site, Point of Contact Agency Information

     Agency                   : Bundesamt fuer Kartographie
                              : und Geodaesie
                              : Geodetical Observatory Wettzell
     Preferred Abbreviation   : BKG
     Mailing Address          : Bundesamt fuer Kartographie
                              : und Geodaesie
                              : Geodetical Observatory Wettzell
                              : Sackenrieder Strasse 25
                              : D-93444 Bad Koetzting
                              : Germany
     Primary Contact
       Contact Name           : Dr. Reiner Dassing
       Telephone (primary)    : ++49-9941-603112
       Telephone (secondary)  : ++49-9941-6030
       Fax                    : ++49-9941-603222
       E-mail                 : [email-masked]
     Secondary Contact
       Contact Name           : Uwe Hessels
       Telephone (primary)    : ++49-9941-603208
       Telephone (secondary)  : ++49-9941-6030
       Fax                    : ++49-9941-603222
       E-mail                 : [email-masked]
     Additional Information   : Klaus Roettcher
                              : ++49-9941-603115
                              : [email-masked]
                              : Rudi Stoeger
                              : ++49-9941-603117
                              : [email-masked]

12.  Responsible Agency (if different from 11.)

     Agency                   : Bundesamt fuer Kartographie
                              : und Geodaesie
     Preferred Abbreviation   : BKG
     Mailing Address          : Bundesamt fuer Kartographie
                              : und Geodaesie
                              : Department Leizig
                              : Karl-Rothe-Strasse 10-14
                              : D-04105 Leipzig
                              : Germany
     Primary Contact
       Contact Name           : Matthias Groeschel
       Telephone (primary)    : ++49-341-5643-432
       Telephone (secondary)  : ++49-341-5643-0
       Fax                    : ++49-341-5643-415
       E-mail                 : [email-masked]
     Secondary Contact
       Contact Name           : Peter Franke (in Frankfurt)
       Telephone (primary)    : ++49-69-6333-279
       Telephone (secondary)  : ++49-69-6333-1
       Fax                    : ++49-69-6333-425
       E-mail                 : [email-masked]
     Additional Information   :

13.  More Information

     Primary Data Center      : BKGI
     Secondary Data Center    : CDDIS
     URL for More Information : www.bkg.bund.de.de  www.fs.wettzell.de
       Hardcopy on File
       Site Map               : (Y or URL)
       Site Diagram           : Y
       Horizon Mask           : Y
       Monument Description   : Y
       Site Pictures          : Y
     Additional Information   : (multiple lines)

     Antenna Graphics with Dimensions

TRM29659.00
>From 2001-07-03 until 2002-04-11

                         -----
                     /     +     \                        <--  0.128  L2
                    |      +      |                       <--  0.110  L1
  +--------------------------------------------------+    <--  0.102  TCR
  |                                                  |
  |                                                  |
  |                                                  |
  |                                                  |
+-+--------------------------------------------------+-+  <--  0.038
+-------------------+-------------+--------------------+  <--  0.035  BCR
                    |             |
                   =|             |
                    +------x------+                       <--  0.000  BPA=ARP

<--                      0.381                       -->

JPSREGANT_SD_E
>From 2002-04-11 until 2005-08-02

                +-------------------------+ <--------------Plastic Housing for
               /    -------------------    \               Antenna/Choke Ring
              /    /         +         \    \                <-- 0.1196 L2
   +---------+     |         +         |     +---------+     <-- 0.1060 L1
  | +-------------------------------------------------+ |    <-- 0.xxxx TCR
 /  |                                                 |  \
|   |                                                 |   |
|   |                                                 |   |
|   |                                                 |   |
 \+-----------------------------------------------------+/
 |+-----------------------------------------------------+|   <-- 0.xxxx BCR
  \______________  |                   |  ______________/
                 \ |                   | /
                  \+---------x---------+/                    <-- 0.0000 BPA=ARP

TRM29659.00
>From 2005-08-02 until 2009-12-16

                         -----
                     /     +     \                        <--  0.128  L2
                    |      +      |                       <--  0.110  L1
  +--------------------------------------------------+    <--  0.102  TCR
  |                                                  |
  |                                                  |
  |                                                  |
  |                                                  |
+-+--------------------------------------------------+-+  <--  0.038
+-------------------+-------------+--------------------+  <--  0.035  BCR
                    |             |
                   =|             |
                    +------x------+                       <--  0.000  BPA=ARP

<--                      0.381                       -->

LEIAR25
>From 2009-12-16

                         +-------+                          <-- 0.1995
                       /     +     \
                      |      +      |
                  |---+-------------+----|		    <-- 0.1614 TCR
              |------------------------------|
          |--------------------------------------|
      |----------------------------------------------|
    +--------------------------------------------------+
    |                                                  |
    |                                                  |
    |                                                  |
  +-+--------------------------------------------------+-+
  +-------------------+-------------+--------------------+  <-- 0.0320 BCR
                      |             |
                      |             |
                      +------x------+                       <-- 0.0000 BPA=ARP

  <--                      0.3802                      -->

   ARP: Antenna Reference Point
   L1 : L1 Phase Center                   L2 : L2 Phase Center
   TCR: Top of Chokering                  BCR: Bottom of Chokering
   TGP: Top of Ground Plane               BGP: Bottom of Ground Plane
   TPA: Top of Preamplifier               BPA: Bottom of Preamplifier
   TOP: Top of Pole

All dimensions are in meters.